The Sentry ISOLOK MSE automatic point sampler is designed for sampling toothpaste, peanut butter, tomato paste, juice concentrates – even rocket propellant. Like any piece of equipment, it sometimes needs repairs and maintenance. But it can be challenging to know how to make strategic decisions around maintaining sampling equipment. Here’s how to keep your ISOLOK MSE sampler working properly.
Seal Replacement challenges
For some operators and maintenance professionals, properly maintaining the specialized parts on the ISOLOK MSE sampler can be challenging.
- The amount of seals within the sampler can cause difficulty in isolating what might be causing errors
- Incorrect placement of seals can cause poor function and allow air into the process stream
- Incorrect seal replacement can cause the sampler to jam – which leads to unreliable sampling
Replacing seals and maintaining equipment can be confusing when just looking at a product manual, and IOMs and spec sheets can provide only so much information.
Learn how to keep your ISOLOK MSE sampler running smoothly with this Seal Replacement and Maintenance How-To video.
This video includes detailed, step-by-step instructions on how to properly maintain the ISOLOK MSE, including:
- What tools are needed to replace and maintain ISOLOK MSE seals
- How to disassemble the automatic point sampler
- How to remove components
- How to remove O-rings from the piston, the sampler body, piston spindle, the sleeve and sleeve bore
- How to install new O-rings onto the sampler body, piston spindle, the sleeve and sleeve bore
- How to reassemble the automatic sampler
Ready to learn more? Watch the How To Video below to properly replace your ISOLOK MSE seals.